Decrease pacer output until capture is lost (typically 0.3 mA or less)
Increase again until capture occurs and to approximately 2.5 times lowest capture threshold
Bessman in Roberts (2014) Roberts and Hedges Emergency Procedures, Elsevier, Philadelphia, p. 277-97
Kwon and Warrington (2016) Crit Dec Emerg Med 30(9):10-11
Orman and Bellezzo in Herbert (2016) EM:Rap 16(4): 8-9
Images: Related links to external sites (from Bing)
These images are a random sampling from a Bing search on the term "Emergency Transvenous Pacing." Click on the image (or right click) to open the source website in a new browser window. Search Bing for all related images
FPnotebook.com is a rapid access, point-of-care medical reference for primary care and emergency clinicians. Started in 1995, this collection now contains 6479 interlinked topic pages divided into a tree of 31 specialty books and 722 chapters. Content is updated monthly with systematic literature reviews and conferences.
Although access to this website is not restricted, the information found here is intended for use by medical providers. Patients should address specific medical concerns with their physicians.
This page was written by Scott Moses, MD, last revised on 9/3/2016 and last published on 3/2/2017.